|Toward imaging the body at 10.5 tesla|
MA Ertürk, X Wu, Y Eryaman, PF Van de Moortele, EJ Auerbach, ...
Magnetic resonance in medicine 77 (1), 434-443, 2017
|Reduction of implant RF heating through modification of transmit coil electric field|
Y Eryaman, B Akin, E Atalar
Magnetic resonance in medicine 65 (5), 1305-1313, 2011
|Parallel transmit pulse design for patients with deep brain stimulation implants|
Y Eryaman, B Guerin, C Akgun, JL Herraiz, A Martin, A Torrado‐Carvajal, ...
Magnetic resonance in medicine 73 (5), 1896-1903, 2015
|In vivo human head MRI at 10.5 T: A radiofrequency safety study and preliminary imaging results|
A Sadeghi‐Tarakameh, L DelaBarre, RL Lagore, A Torrado‐Carvajal, ...
Magnetic resonance in medicine 84 (1), 484-496, 2020
|Reduction of the radiofrequency heating of metallic devices using a dual‐drive birdcage coil|
Y Eryaman, EA Turk, C Oto, O Algin, E Atalar
Magnetic resonance in medicine 69 (3), 845-852, 2013
|Transcranial focused ultrasound for BOLD fMRI signal modulation in humans|
L Ai, JK Mueller, A Grant, Y Eryaman, W Legon
2016 38th Annual International Conference of the IEEE Engineering in …, 2016
|Fast electromagnetic analysis of MRI transmit RF coils based on accelerated integral equation methods|
JF Villena, AG Polimeridis, Y Eryaman, E Adalsteinsson, LL Wald, ...
IEEE Transactions on Biomedical Engineering 63 (11), 2250-2261, 2016
|First in‐vivo human imaging at 10.5 T: Imaging the body at 447 MHz|
X He, MA Ertürk, A Grant, X Wu, RL Lagore, L DelaBarre, Y Eryaman, ...
Magnetic resonance in medicine 84 (1), 289-303, 2020
|Miniaturized fiber‐optic transmission system for MRI signals|
OG Memis, Y Eryaman, O Aytur, E Atalar
Magnetic Resonance in Medicine: An Official Journal of the International …, 2008
|SAR reduction in 7T C‐spine imaging using a “dark modes” transmit array strategy|
Y Eryaman, B Guerin, B Keil, A Mareyam, JL Herraiz, RK Kosior, A Martin, ...
Magnetic resonance in medicine 73 (4), 1533-1539, 2015
|Magnetic resonance method and apparatus for reducing RF heating in the patient|
Y Eryaman, E Atalar
US Patent 8,319,496, 2012
|A simple geometric analysis method for measuring and mitigating RF induced currents on Deep Brain Stimulation leads by multichannel transmission/reception|
Y Eryaman, N Kobayashi, S Moen, J Aman, A Grant, JT Vaughan, ...
Neuroimage 184, 658-668, 2019
|Evaluation of internal MRI coils using ultimate intrinsic SNR|
H Çelik, Y Eryaman, A Altıntaş, IA Abdel‐Hafez, E Atalar
Magnetic Resonance in Medicine: An Official Journal of the International …, 2004
|Multi‐atlas and label fusion approach for patient‐specific MRI based skull estimation|
A Torrado‐Carvajal, JL Herraiz, JA Hernandez‐Tamames, ...
Magnetic resonance in medicine 75 (4), 1797-1807, 2016
|10.5 T MRI static field effects on human cognitive, vestibular, and physiological function|
A Grant, GJ Metzger, PF Van de Moortele, G Adriany, C Olman, L Zhang, ...
Magnetic resonance imaging 73, 163-176, 2020
|Improving radiofrequency power and specific absorption rate management with bumped transmit elements in ultra‐high field MRI|
A Sadeghi‐Tarakameh, G Adriany, GJ Metzger, RL Lagore, S Jungst, ...
Magnetic resonance in medicine 84 (6), 3485-3493, 2020
|Design of internal MRI coils using ultimate intrinsic SNR|
Y Eryaman, Y Öner, E Atalar
Magnetic Resonance Materials in Physics, Biology and Medicine 22, 221-228, 2009
|Combined loop+ dipole arrays for 7 T brain imaging|
Y Eryaman, B Guerin, R Kosior, E Adalsteinsson, LL Wald
Proc 21st Annual Meeting ISMRM, Salt Lake City, 48, 2013
|A new coil element for highly-dense transmit arrays: An introduction to non-uniform dielectric substrate (NODES) antenna|
A Sadeghi-Tarakameh, S Jungst, X Wu, M Lanagan, G Adriany, ...
Proceedings of the 27th Annual Meeting of ISMRM, Montreal, Canada 732, 2019
|Radiofrequency heating studies on anesthetized swine using fractionated dipole antennas at 10.5 T|
Y Eryaman, RL Lagore, MA Ertürk, L Utecht, P Zhang, A Torrado‐Carvajal, ...
Magnetic resonance in medicine 79 (1), 479-488, 2018